Non-invasive Level Gauge
Max Measuring Range 30meters (100ft)
No downtime/empty tank during installation
No tank break in; Non-drilling; No flange
100% safe, no leakage risk
No moving part, No maintenance
Operated by a remote control, No PC, No Oscilloscope
Local display + Remote output (4-20mA/RS-485)
Best for flammable, toxic, vaporized liquid.
Whether your tank management application is best served by Radar, Servo, or Ultrasonic, SKE offers advanced level measurement solutions:
– Non-invasive Sensing Technique
This Non-invasive Level Gauge adopts the principle of Sonar Ranging.
When the probe sensor is bonded to the bottom wall of the tank, a sound wave travels through the tank/medium and reflects back from the liquid surface.
The sensor is responsible for emitting and receiving sound waves, then transmits them to the system, and calculates the height (H) of the liquid, from the time (t) taken for the signal to be received.
SKE self-developed probe sensor applies piezoelectric ceramic materials.
The magnetic ring is setting around the sensor. Pretty easy to install on a ferromagnetic material tank (exp: carbon steel).
For a non-ferromagnetic material tank(exp: stainless), the steel adapter board with the package would help to fix it.
According to different working conditions, measuring range, tank type/size, there are three types of probe sensors with different sizes and powers to be selected
Measurement (MEA) & Calibration(CAL, TEMP)
Generally, a tank monitoring case often requires two probe sensors at a time:
– The one on the bottom for level measurement, called Measurement Sensor (MEA)
– The other is the Diameter Calibration Sensor(CAL), which bonded on the sidewall, to continuously calibrate the real-time sound velocity and update to the Transmitter.
In some practical applications, the sound velocity would be varied due to environment/liquid temperature change. The accurate sound velocity value would be a key point for the level measurement result:
When the level inside is lower than the CAL sensor position, Diameter Calibration would be off work.
At that time, Temperature Calibration(TEMP) would be automatically turned on. It keeps recording the temperature change, transmits data to the software algorithm, then obtains accurate measurement results.
Two Wire Non-Intrusive Ultrasonic Level Gauge
The Non-invasive Level Gauge is in use everywhere where the level of liquids in tanks has to be measured continuously and contactlessly. Application possibilities for radar gauges can be found in almost all areas of industry. Sewage treatment plants and water purification plants in particular, but also other environmental engineering facilities as well as power generating stations now rely on cost-effective radar sensors for continuous level measurement.
In general, the dynamic viscosity of the measured medium is required to be less than 10 MPa. S. When 10 MPa. S < dynamic viscosity < 30 MPa. S, the measure range may be reduced and the blind area may be increased. It cannot be measured when the dynamic viscosity is > 30 MPa. S. (1 MPa. S = 1 CP). There should be no dense bubbles in the liquid; there should not be a large number of suspended substances in the liquid, such as crystals; there should not be a large number of sedimentary substances in the liquid, such as mud and sand. The temperature of the measured medium shall be within – 50 ℃ ~ 100 ℃.
|Liquid ammonia||Methyl alcohol||Methylbenzene||Butadiene|
|98% sulfuric acid||Diesel||Gasoline||Naphtha|
|Hydrogen fluoride||Hydrofluoric acid||Butane||Hydroxy nickel|
|Max Measuring range||30 m|
|Signal output||(4~20 )mA, HART,MAX 500Ω|
|Environment temperature in using lever gauge||-20℃~+70℃,-40℃~+70℃|
|Environment temperature in using probe||-50℃~+100℃,60℃~330℃|
|Explosion-proof sign||Exd II CT6|
|Enclosure protection grade||IP67|
|Power supply||DC 24V|
|The cable length from probe to main engine||5m,10m,15m,20m,25m,30m|
|Weight of mainframe||2.5KG|
|Size of mainframe||217mm(length)*160mm(width)*185mm(height)|
|The diameter of the base hole in mainframe||6mm|